DescriptionRT11-0.4-70 Stationary anode X-Ray Tube is especially designed for high frequency intra-oral dental x-ray unit and available for a nominal tube voltage with DC. RT11-0.4-70 tube has one focus. The integrated high quality tube with glass design has one super imposed focal spot and a reinforced anode. The high anode heat storage capacity ensures a wide range of applications for intra-oral dental application. A special designed anode enables an elevated heat dissipation rate which leads to a higher patient throughput and a longer product life. A constant high dose yield during the entire tube life is ensured by the high density tungsten target. Ease of integration into system products is facilitated by extensive technical support.
|Maximum Tube Voltage||70kV|
|Maximum power (at 1.0s)||430 W|
|Maximum anode cooling Rate||110W|
|Max.Anode Heat Content||4.3kJ|
|Focal spot||0.4(IEC 60336 2005)|
|Cathode type||W filament|
|Permanent Filtration||Min. 0.5mmAl/50 kV(IEC60522/1999)|
|Dimensions||67mm length by 30mm diameter|
Competitive AdvantageElevated anode heat storage capacity and cooling Constant high dose yield Excellent lifetime
ApplicationsRT11-0.4-70 Stationary anode X-Ray Tube is especially designed for high frequency intra-oral dental x-ray unit and available for a nominal tube voltage with DC.
Maintaining seasoning schedule Prior to usage, season the tube in accordance with the seasoning schedule given below until the required tube voltage is attained. Example given – needs to revise by the manufacturer and specified in the data sheet of the part: Initial incoming seasoning and seasoning schedule for idle period (more than 6 months) Circuit: AC/DC (Center grounded) When the tube current is unstable in seasoning, immediately switch off the tube voltage and after an interval of 5 minutes or more, increase the tube voltage gradually from the low voltage while making sure that the tube current is stable. The withstand voltage performance of the tube unit will be lowered as the exposure time and the number of operation increase. Stain-like impact traces might appear on the x-ray tube target surface by slight discharge in the course of the seasoning. These phenomena are one process to recover the withstanding voltage performance at that time. Therefore, if it is in stable operation at the maximum tube voltage of seasoning subsequent to them, the tube unit can be used without any interference to its electrical performance which in use.